Fill out the form to get a free trial:
1Department of Microbiology and Immunology, Lineberger Comprehensive Cancer Center, Center for AIDS Research, University of North Carolina at Chapel Hill
Fill out the form to get a free trial:
We will demonstrate the setup and analysis of pre-microRNA 96-well arrays for QPCR using a robot as well as by hand with a Thermo Scientific Matrix multichannel pipette.
Chugh, P., Tamburro, K., Dittmer, D. P. Profiling of Pre-micro RNAs and microRNAs using Quantitative Real-time PCR (qPCR) Arrays. J. Vis. Exp. (46), e2210, doi:10.3791/2210 (2010).
The qPCR pre-miRNA profiling arrays can be set up as fully automated with a Tecan Freedom Evo robot (A) or by hand with the Matrix electronic multichannel pipette (B).
1)Prepare the master mix, primer plates and samples.
A. Setup of pre-miRNA assay using the Freedom Tecan Evo robot.
B.Setup of pre-miRNA assay using the Matrix electronic multichannel pipette:
Secrets to success:
An important characteristic when designing PCR arrays is that all 186 primer pairs have the same annealing temperature, so the plate can be run at the same optimal PCR run program settings.
Each new array should be checked using three controls prior to running samples. These controls are:
1 run of all primers run with water or .1x TE to rule out primer contamination.
1 run of alternating water/TE and positive control, to ensure no carryover.
3 runs using the same positive control, to ensure reproducibility between runs.
Master mixes should be freshly prepared, and plates should be run within 24 hrs for optimal results
Primer plate foil should be removed slowly to reduce risk of contamination between wells.
When using a robot with fixed tips, 2% bleach wash is necessary to wash the tips between each pipetting step, followed by a water flush. This prevents and eliminates carryover, which could otherwise contaminate data and lead to inconclusive results.
After a plate is run through the Lightcycler, it should not be opened again in the PCR setup room. This helps prevent PCR contamination.
qPCR results are typically represented by CT values as determined from the Lightcycler analysis software. A successful run usually consists of a range of CTs for samples, typically between 20-35 for samples that are positive. The water samples in a good run are always at a CT of >40 and any samples or specific wells with a CT over 37 are considered negative or not detected. As a general rule, samples yielding a CT of <10 are also unreliable and are excluded from further analysis. There are several ways to analyze qPCR data and the inclusion of internal controls in our assay helps control for variation of sample input. The pre-miR array includes a U6 control primer, which is often used as a reference gene to normalize CT values from different samples. This normalized value is referred to as the delta CT value (dCT). Results are often graphed as the raw CT, the dCT value or can be further analyzed to standardized values.
In Figure 1, analysis of the full pre-microRNA array of four different samples from a timecourse experiment is presented in heat map format. The relative expression of each pre-microRNA is shown and three major, distinct clusters emerged. The majority of pre-miRs analyzed underwent small, insignificant changes as expected (Figure 1A). However, a small portion of pre-microRNAs were significantly decreased (Figure 1B) or increased dramatically (Figure 1C) throughout the timecourse experiment. Levels of expression of each pre-microRNA were normalized to the 0h timepoint (sample 1) as a baseline level. In some instances, you may note that some of the clustered pre-microRNAs also cluster together in the heatmap analysis (ex: let-7a, Figure 1B). Depending on the experiment, clusters may emerge that are dependent on viral infection, cell type specific expression of pre-microRNAs or pre-miRs that are regulated by a common molecule or signaling pathway.
The pre-microRNA assay that we have developed also includes a number of known viral pre-microRNAs and genes encoded by Kaposi's Sarcoma Associated Herpesvirus (KSHV) and Epstein Barr Virus (EBV) in addition to the human pre-miRs. These can be used as positive or negative controls, depending on the cell line and viral status of the samples used. In Figure 2, the average CTs from a KSHV-negative sample run in quadruplicate are shown. Importantly, KSHV LANA was not detected in this sample by the highly sensitive qPCR array. However, U6--our internal control and let-7a--a highly expressed human pre-microRNA, were expressed at detectable levels. Finally, as expected, the negative control of PCR-grade water did not yield a qPCR product.
Another indicator of a successful qPCR run is a good melting curve analysis, which can give insight into potential contamination in samples. To establish a melting temperature, the plate is heated slowly after the PCR is complete. As the DNA strands separate, SYBR green is released, and fluorescence signal decreases. The temperature at which this drop occurs is graphed, and is known as the melting temperature of the sample. Depending on the DNA sequence, samples melt at different temperatures from each other and from the water negative control, which should not have a melting temperature since no DNA is present. The PCR products from each primer pair should melt at similar temperatures. For example, Figure 3 shows a melting curve analysis for two different primers using the same sample in duplicate. It is clear that the melting temperature is different for the two primer pairs but the sample is melting at the exact same temperature for each replicate. Signs of a bad or potentially contaminated sample may include several melting peaks, suggesting the presence of two different sources of input.
Figure 1. Pre-microRNA signatures emerge through profiling using a novel qPCR-based array. The average deltaCTU6 was calculated and standardized values were loaded into ArrayMinerTM software, yielding 3 distinct clusters displayed as heatmaps. (A) Pre-microRNAs with small changes in expression are shown. Pre-microRNAs whose levels were significantly decreased (B) or increased (C) are also shown. Blue corresponds to lower levels of expression while red represents increased expression levels.
Figure 2. Inclusion of internal controls in the qPCR-based pre-microRNA array. The raw CT values for 3 different genes and a no template control are shown. U6 is used as an internal positive control while PCR H2O serves as our negative control. Let-7a-1-1 is a microRNA normally expressed in many different cell types while KSHV LANA can be used as either a positive or negative control, based on viral status of the cell line or tissue used.
Figure 3. Melting peak analysis of pre-microRNA primers and lack of sample contamination. Melting curves were obtained using the Tm calling analysis in the LightCycler software. Two different primers (Primer A and B) for the same sample run in duplicate are shown. The melting temperature for each primer is distinct. However, the sample has only one peak, demonstrating the lack of sample contamination.
QPCR is a highly sensitive assay that can be used to compare gene expression levels between samples in a study, as well as to determine positivity in the case of viruses. The benefit of using qPCR arrays is the ability to run many primers (in our case, 186 primer pairs) for each sample in a short period of time. Using a pipetting robot such as the Tecan Freedom Evo, the amount of time required to set an experiment up can be further reduced, and the accuracy and consistency of the robot can reduce and eliminate pipetting errors. Here we demonstrate how to use a Tecan Freedom Evo to set up a microRNA array, as well as an alternative method using a Matrix Electronic Multichannel Pipette, which can be implemented for labs who do not have access to a pipetting robot. qPCR arrays are also beneficial because all genes can be normalized to a single set of reference genes run on the same plate, reducing the cost and number of reactions.
As mentioned, there are several key features that are necessary when designing such an array. Since all samples on the PCR plate will be run at identical conditions, it is critical to design primers that function at the same temperatures and settings, or some primer pairs will fail. When designing primers, the sequences should always be checked by BLAST search to ensure that there is homology to your gene of interest and no other genes.
When running the array, it is critical that the environment is sterile and free of PCR contaminants that may result in false positives. If possible, PCR should be set up in a sterile work hood or a separate clean room, where PCR products are never used. Benches should be washed with Eliminase, water, and ethanol, and subjected to UV light for at least an hour after the run is complete.
The benefit of this technique is that this same strategy is applicable for any set of genes to be profiled by simply replacing the miRNA array with an array containing a different primer pair set. Our lab currently contains arrays for pre- miRNAs, KSHV, EBV, p53, and NFkB - all of which can be run in an identical method to those shown. The flexibility of this assay allows our lab to perform high-throughput gene profiling rapidly and reliably for any primer pairs that we wish to screen for.
Here, we described the array using cDNA from human cell lines to profile miRNA expression levels upon activation over a period of time. In addition to characterizing cell lines, the array can also be used to analyze clinical samples. Following RNA isolation, cDNA synthesis can be performed, and the cDNA can be run using this array to characterize gene expression. This can be useful in determining viral positivity, or gene expression profiles to further characterize which genes are upregulated in samples from infected patients. In summary, automated qPCR-based arrays are reproducible and reliable assays that allow for highly sensitive detection of a wide variety of genes from tissues and cell lines in a timely fashion.
No conflicts of interest declared.
This work was supported by NIH grants DE018304, R01DE018281. K.T. is supported by T32 GM07092-34 and by a grant to the University of North Carolina at Chapel Hill from Howard Hughes Medical Institute (HHMI) through the Med into Grad Initiative. PC is supported by T32 CA009156.
|Freedom Evo 150||Tecan Group Ltd.||30017587|
|Matrix Electronic Multichannel Pipette||Thermo Fisher Scientific, Inc.||2001-MTX||(or any comparable Thermo Scientific multichannel that can pipette the volumes described above)|
|Matrix Integrity Filter Tips, Sterile||Thermo Fisher Scientific, Inc.||7435||(or comparable tip for multichannel used)|
|Lightcycler 480 SYBR Green 1 Master||Roche Group||04 707 516 001|
|Lightcycler 480 Instrument II||Roche Group||05015243001||384-well version|
|Lightcycler 480 Multiwell Plate 384, white||Roche Group||04729749001|
|LightCycler 480 Sealing Foil||Roche Group||04729757001|
|LightCycler 480 Multiple Plate Analysis Software||Roche Group||05075122001|
|Eliminase Decontaminant||Decon Laboratories||04-355-31|
You must be signed in to post a comment. Please sign in or create an account.