Fill out the form to get a free trial:
1Children's Mercy Hospital and Clinics, School of Medicine, University of Missouri-Kansas City
Fill out the form to get a free trial:
This protocol presents a complete and detailed procedure to apply RNA-seq, a powerful next-generation DNA sequencing technology, to profile transcriptomes in human pulmonary microvascular endothelial cells with or without thrombin treatment. This protocol is generalizable to various cells or tissues affected by different reagents or disease states.
Keywords: Genetics, Issue 72, Molecular Biology, Immunology, Medicine, Genomics, Proteins, RNA-seq, Next Generation DNA Sequencing, Transcriptome, Transcription, Thrombin, Endothelial cells, high-throughput, DNA, genomic DNA, RT-PCR, PCR
Cheranova, D., Gibson, M., Chaudhary, S., Zhang, L. Q., Heruth, D. P., Grigoryev, D. N., et al. RNA-seq Analysis of Transcriptomes in Thrombin-treated and Control Human Pulmonary Microvascular Endothelial Cells. J. Vis. Exp. (72), e4393, doi:10.3791/4393 (2013).
The characterization of gene expression in cells via measurement of mRNA levels is a useful tool in determining how the transcriptional machinery of the cell is affected by external signals (e.g. drug treatment), or how cells differ between a healthy state and a diseased state. With the advent and continuous refinement of next-generation DNA sequencing technology, RNA-sequencing (RNA-seq) has become an increasingly popular method of transcriptome analysis to catalog all species of transcripts, to determine the transcriptional structure of all expressed genes and to quantify the changing expression levels of the total set of transcripts in a given cell, tissue or organism1,2 . RNA-seq is gradually replacing DNA microarrays as a preferred method for transcriptome analysis because it has the advantages of profiling a complete transcriptome, providing a digital type datum (copy number of any transcript) and not relying on any known genomic sequence3.
Here, we present a complete and detailed protocol to apply RNA-seq to profile transcriptomes in human pulmonary microvascular endothelial cells with or without thrombin treatment. This protocol is based on our recent published study entitled "RNA-seq Reveals Novel Transcriptome of Genes and Their Isoforms in Human Pulmonary Microvascular Endothelial Cells Treated with Thrombin,"4 in which we successfully performed the first complete transcriptome analysis of human pulmonary microvascular endothelial cells treated with thrombin using RNA-seq. It yielded unprecedented resources for further experimentation to gain insights into molecular mechanisms underlying thrombin-mediated endothelial dysfunction in the pathogenesis of inflammatory conditions, cancer, diabetes, and coronary heart disease, and provides potential new leads for therapeutic targets to those diseases.
The descriptive text of this protocol is divided into four parts. The first part describes the treatment of human pulmonary microvascular endothelial cells with thrombin and RNA isolation, quality analysis and quantification. The second part describes library construction and sequencing. The third part describes the data analysis. The fourth part describes an RT-PCR validation assay. Representative results of several key steps are displayed. Useful tips or precautions to boost success in key steps are provided in the Discussion section. Although this protocol uses human pulmonary microvascular endothelial cells treated with thrombin, it can be generalized to profile transcriptomes in both mammalian and non-mammalian cells and in tissues treated with different stimuli or inhibitors, or to compare transcriptomes in cells or tissues between a healthy state and a disease state.
A flowchart outlining this protocol is displayed in Figure 1.
1. Treatment of Cells with Thrombin, RNA Isolation, Quality Assessment and Quantification of RNA
2. Library Construction and Sequencing
3. Data Analysis
4. Validation of the RNA-seq Results by Quantitative Real-time-Polymerase Chain Reaction (qRT-PCR)
For Step 1: The 28s:18s ratio is traditionally used as an indicator of RNA degradation. Ideally, the 28s peak should have approximately twice the area of the 18s band (a ratio of 2), however this ideal ratio is often not seen in practice. Furthermore, 28s:18s ratios obtained from spectrophotometric methods can underestimate the amount of degradation of the RNA. To more accurately quantify the degradation, and therefore the quality of the RNA sample, the Experion system calculates an RNA Quality Indicator (RQI) number. The RQI algorithm compares the electropherogram of RNA samples to data from a series of standardized, degraded RNA samples and automatically returns a number between 10 (intact RNA) and 1 (degraded RNA). The RNA quality should have an RQI of at least 7, ideally greater than 8. Figure 2 shows the Experion results using a high-quality RNA sample with an RQI of 8.4.
For Step 2: The libraries should have a broad band at approximately 250-300 bp. Figure 3 shows Experion results of a high-quality library. Figure 4 shows the qPCR results of standard curve samples and one unknown sample (shown in dark blue). The progress and quality of the sequencing run should be constantly observed throughout the run. Figure 5 shows appropriate cluster density during the first cycle imaging step; this is the first indication of the run quality. Clusters should be bright and focused. Figure 6 shows the First Base Report generated after the first cycle is complete. It is important to assess the estimated cluster density, intensity levels, and focus quality at this point. The next quality checkpoint, after cycle 4, is shown in Figure 7. This shows the absolute cluster density for each lane. The cluster density should not be above 850 k/mm2. After cycle 13, phasing (when a base is not added during a cycle) and prephasing (when two bases are added during a cycle) stats are calculated, as shown in Figure 8. Typical numbers are between 0.1 and 0.25. The major quality assessment is possible after cycle 24, when several quality metrics are calculated. The percent of reads above Q30, shown in Figure 9, is a measure of the confidence in the base calling. A read with a Q score of 30 means that there is a 1 in 1,000 chance that base call is wrong. The Q scores will decrease as the run progresses, but should start out with greater than 95% of the reads meeting or exceeding Q30. The clusters passing filter (PF), shown in Figure 10, are the clusters from which the actual sequence data will be taken. Ideally, this should be above 85%. The cluster PF is based on many factors, including phasing, prephasing, intensity and Q30. It will not change as the run progresses. The percent aligned (Figure 11) is a measure of the reads that align real-time to the PhiX genome. Since we spiked in approximately 1% PhiX library to the sample libraries, the percent aligned should be between 0.5 and 1. This statistic shows that the library content is represented well by the clusters and there was no cluster generation bias.
For Step 3: Table 1 presents expressed genes and isoforms in both control and thrombin-treated human pulmonary microvascular endothelial cells. Notably, there are about 26,000 novel isoforms detected, which illustrates the strength of RNA-seq—it can identify unknown RNAs, alternatively spliced transcripts and alternative promoter usage which are not detectable by microarray techniques3. RNA-seq can also measure the less abundant transcripts that are inaccurately quantified or not detected by microarrays. Figure 12 and Table 2 display differentially expressed genes in the thrombin signaling pathway. This is an example of the third generation knowledge base-driven pathway analysis9: pathway topology based approaches, using Ingenuity Pathway Analysis software. It shows that a six h thrombin treatment significantly up-regulates the thrombin receptor Par 4 and down-regulates the thrombin receptor Par 3 while there is no change in the expression of the thrombin receptor Par 1. Expression of NFkB1, NF kB2 and Src are also significantly up-regulated. Some of Rho family genes (Rho B, C, F and G) are up-regulated while others (Rho J, Q, T1, U and V) are down-regulated (Table 4). Myosin light chain gene 9 (MYL9) is up-regulated while MYL12B is down-regulated. Expression of other genes is either down-regulated or not affected.
For Step 4: To validate the RNA-seq results using an alternative approach, we performed a qRT-PCR experiment to assay three different genes (Figure 13). In RNA-seq data, TRAF1 was up-regulated by 7.96 fold; CELF1 was down-regulated by 1.16 fold; and FANCD2 was down-regulated by 1.70 fold. In qRT-PCR data, these corresponding numbers are +7.25 fold, -1.15 fold and -2.07 fold, respectively. The results of these three genes assayed by RNA-seq and qRT-PCR are in good agreement, which corroborates the RNA-seq results.
|Total Genes Expressed||16,636||16,357|
|Up-regulated (2-fold or greater difference)||152|
|Down-regulated (2-fold or greater difference)||2,190|
|Total Known Isoforms Expressed||26,807||26,300|
|Up-regulated (2-fold or greater difference)||480|
|Down-regulated (2-fold or greater difference)||3,574|
|Total Novel Isoforms Expressed||25,880||25,886|
|Up-regulated (2-fold or greater difference)||1,775|
|Down-regulated (2-fold or greater difference)||12,202|
Table 1. Gene/Isoform Expression Summary*. This table lists genes, known isoforms and novel isoforms expressed in control and thrombin-treated HMVEC-LBl cells. The fold change is the ratio of thrombin FragmentsPerKilobase of transcript perMillion fragments mapped (FPKM) to control FPKM. Those genes, known isoforms and novel isoforms with 2-fold or greater difference between two groups have statistically different expression levels (as determined after Benjamini-Hochberg correction). * This table is reproduced from Table 2 in Reference 4.
|Gene Symbol||Overall Fold Change||Members||Individual Fold Change||q_value**||FPKM Control||FPKM Thrombin|
|Symbol||Overall Fold Change||Members||Individual Fold Change||q_value**||FPKM Control||FPKM Thrombin|
|Partial Up-/Partial Down-Regulated Genes|
|Symbol||Overall Fold Change||Members||Individual Fold Change||q_value**||FPKM Control||FPKM Thrombin|
Table 2. Differentially Expressed Genes and Isoforms in Thrombin Signaling Pathway*. *, This table is reproduced from Table S4 in Reference 4 with a minor modification. **, q-value, a false discovery rate adjusted p-value.
Figure 1. Flowchart of the protocol for RNA-seq profiling of the thrombin-mediated transcriptome in human pulmonary microvascular endothelial cells. First, treat cells and isolate and quantify RNA. Prepare libraries from the high-quality RNA and assess their concentration (via qPCR) and quality (via a bioanalyzer), then cluster on a flow cell. Start the sequencing run and analyze the quality of the run throughout. The major quality checkpoints are cycles 1, 4, 13 and 24. Using CASAVA, convert the bcl files to fastq files and then align those to the genome with TopHat. Detect known and unknown isoforms using Cufflinks and determine differential expression with CuffDiff. View the output files in Excel and filter out genes that have an expression level less than 0.05 FPKM or a p-value larger than 0.05. Submit the remaining genes to Ingenuity Pathway Analysis for further information about the gene functions and pathways affected by the thrombin treatment. Usually, selected set of differentially expressed genes are validated by an alternative approach such as qRT-PCR.
Figure 2. A representative RNA sample with an RQI of 8.4 as analyzed by the Experion Automated Electrophoresis Station. A) The individual trace of high-quality RNA - the x-axis depicts time and the y-axis depicts fluorescent signal. B) The virtual gel picture of a high-quality RNA sample. The intensity of the 28S peak is greater than 18S peak and no contamination is seen. Both bands are sharp and defined. Click here to view larger figure.
Figure 3. High quality library prepared from RNA as analyzed by the Experion Automated Electrophoresis Station. A broad peak between 250 and 300 bp is detected and no high molecular weight DNA (contaminating DNA) is observed. A) The individual trace of a high-quality library - the x-axis depicts time and the y-axis depicts fluorescent signal B) The virtual gel picture of a high-quality library. A broad peak between 250 and 300 bp is detected and no high molecular weight DNA (contaminating DNA) is observed. Click here to view larger figure.
Figure 4. qPCR results using SyberGreen and primers specific to Illumina ligated adapters. A previously clustered library is used as standard curve to determine the optimal clustering concentration for the newly prepared library. The dilution of the unknown library falls within the standard curve concentrations The standard curve samples are indicted by the arrows and the unknown sample is dark blue and also indicated by an arrow.
Figure 5. Appropriate cluster density during the first cycle imaging step. The clusters are clear, focused and bright. A) Base A; B) Base C; C) Base G; D) Base T.
|Metric||Lane 1||Lane 2||Lane 3||Lane 4||Lane 5||Lane 6||Lane 7||Lane 8|
|Cluster Density (k/mm2)||420.94||412.95||410.81||408.54||416.71||416.56||410.02||411.84|
|A Focus Score||68.2||67.46||67.67||67.75||67.41||67.43||67.63||67.05|
|C Focus Score||67.93||67.33||67.56||67.66||67.33||67.39||67.46||66.9|
|G Focus Score||65.4||64.14||63.98||63.92||63.29||63.41||63.47||63.74|
|T Focus Score||66.45||65.57||65.5||65.49||65.03||65.23||65.57||65.59|
Figure 6. The First Base Report generated after the first cycle is complete. The cluster density (although underestimated at this stage) is appropriate. The intensities look good (10,000-26,000, with G having the lowest intensity). The focus scores are also appropriate, falling around 65-70, although higher values are also appropriate.
Figure 7. The cluster density calculated after cycle 4. The cluster density is lower than 850 k/mm2 and even across all lanes. A) Tabular form; B) Graph form. Click here to view larger figure.
Figure 8. Phasing (not adding a base during a cycle) and Pre-phasing (adding two bases during a cycle). Both values are at 0.25 or below, indicating appropriate phasing/pre-phasing levels. A) The phasing and pre-phasing numerical values. B) The phasing values in graph form. C) The pre-phasing values in graph form. Click here to view larger figure.
Figure 9. The different representations of Q30 scores after cycle 24. A score of 30 or higher indicates a 1 in 1,000 chance that base call is wrong. Around 90% of the Q scores are above 30. A) tabular form (Q30 scores here are from the entirety of the run, through cycle 24); B) Q30 scores by cycle. Each bar represents the distribution of the reads falling at Q30 or above for that particular cycle. C) Q30 score distributions through cycle 24. The distribution of Q scores on a cumulative basis through cycle 24. Q score is on the x-axis and millions of reads is on the y-axis. Reads with a Q30 above 30 are represented in green. Click here to view larger figure.
Figure 10. The different representations of clusters passing filter. Clusters passing filter is based on several parameters, including Q30, intensity and phasing/pre-phasing. At least 85% of the clusters are passing filter in each lane. A) Tabular form; B) Graph form, blue boxes represent the total number of clusters, green boxes represent the clusters passing filters. Click here to view larger figure.
Figure 11. The percent of the clusters aligned to the PhiX genome. Approximately 0.5% of the clusters align to the PhiX genome, appropriate for the amount of PhiX spiked into the samples. A) Tabular form; B) Graph form. Click here to view larger figure.
Figure 12. Genes differentially expressed by the thrombin treatment of HMVEC-LBl in the Thrombin Signaling pathway. The Thrombin Signaling pathway was constructed by Ingenuity. In the pathway, red color indicates up-regulated genes(3 1.3 fold) and green down-regulated genes (3 1.3 fold) by the thrombin treatment while white indicates no change of gene expression compared to the control cells. The detailed fold changes and full name of each gene or isoform are presented in Table 2. This figure is reproduced from Figure 4 in Reference 4. Click here to view larger figure
Figure 13. qRT-PCR validation of three differentially expressed genes from thrombin-treated HMVEC-LBl RNA-seq data. qRT-PCR was carried out as described in the protocol. Fold changes determined from the relative Ct values of the TaqMan Gene Expression assay for CUGBP, Elav-like family member 1 (CELF1), Fanconi anemia, complementation group D2 (FANCD2) and TNF receptor-associated factor 1 (TRAF1) were compared to those detected by RNA-seq. Replicates (n=4) of each sample were run and the Ct values averaged. All Ct values were normalized to β-actin. The error bars represent the range of the fold change as determined by the Data Assist software. p<0.05 was considered statistically significant in relative fold changes between thrombin-treat group and control group by both the RNA-seq and the qRT-PCR assays. The mRNA level in control groups by each assay was arbitrarily set as one, which are not shown. *, p<0.05; **, p<0.01. This figure is reproduced from Figure 6 in Reference 4.
RNA Handling: RNases will degrade even the most high-quality RNA, therefore care must be taken during the isolation, storage and use of RNA10. Gloves are always worn to prevent contamination by RNases found on human hands. Gloves should be changed often, particularly after touching skin, doorknobs or other common surfaces. A set of pipettes should be dedicated solely to RNA work and all tips and tubes should be RNase-free. RNA isolation and downstream application should be performed in low-traffic areas that are routinely decontaminated with a product such as RNase-Zap. Degradation can also occur with repeated freeze-thaw cycles. These can be minimized by aliquoting RNA for downstream applications immediately upon isolation. Alternatively, cDNA can be made for downstream application such as qRT-PCR directly after isolation. RNA should be stored at -80 °C and kept on ice during use. It is also important to thoroughly thaw and vortex RNA samples before and during use.
The starting quality of the RNA is essential to the success of this protocol. Use of degraded or otherwise low-quality RNA can cause the library preparation to fail or result in low yield that will be insufficient for sequencing. Even if enough library can be made from degraded or low-quality RNA, it will not accurately represent the transcripts present in the sample, resulting in inaccurate quantification of expression. Also, any unremoved genomic DNA during the RNA isolation can cause an over-estimation of the amount of RNA used in the protocol, but if an amount (e.g. 1-2 μg) in the middle of the suggested range (0.1-4 μg) is used, the actual amount of RNA will be sufficient for the protocol. In addition, any genomic DNA is not likely to be picked up by oligo(dt) primer based library construction in the standard Illumina protocol and thus it will not cause downstream issues with expressed mRNA transcript levels. The quantification of the starting RNA using regular spectrophotometric readings should be sufficient and there is no need to use highly accurate quantitative methods such as RiboGreen since the libraries will be accurately quantified after preparation by qPCR and gene expression levels are normalized relative to the total number of reads during the data analysis steps.
Library Quantification. Accurate quantification of the libraries is vital to the appropriate generation of clusters. Only DNA molecules with successfully ligated adapters will form clusters. Spectrophotometric readings will not differentiate between DNA with adapters and DNA without, which will result in an over-estimation of the concentration of the library and ultimately result in under-clustering of the flow cell. The amount of DNA without ligated adapters will differ from library to library, even if the same RNA was used as starting material, so one cannot accurately assume that a standard percentage of the spectrophotometric reading is DNA with ligated adapters. Performing qPCR with primers specific to the adapters will detect only those DNA molecules with adapters successfully ligated to both ends of the molecule. This allows an absolute quantification of the libraries and in turn an accurate cluster density. Assessment of the libraries with an Experion instrument or a bioanalyzer is also important. This allows a visualization of the size range of the library, which is important during the data analysis steps, and ensures that there is no contamination that may interfere with cluster generation.
Data Analysis. In this protocol, we applied TopHat to align RNA-seq reads to human reference transcriptome and Cufflinks to assemble transcripts, estimate their abundances, and test for differential expression in thrombin-treated HMVEC-LBl cells. TopHat and Cufflinks are two popular tools, and they are free, open-source software11. A recent study showed that Cufflinks had a high sensitivity and specificity in detecting previously annotated introns12. However, TopHat and Cufflinks do not address all applications of RNA-seq nor are they the only tools for RNA-seq analysis11. TopHat and Cufflinks require a sequenced reference transcriptome or genome. They are particularly suitable for the analysis of RNA-seq data generated from either Illumina or SOLiD sequencing machines but not 454 or the classic capillary electrophoresis sequencing platform. Users working without a sequenced reference genome may consider performing de novo transcriptome assembly using one of several tools such as Trinity (http://trinityrnaseq.sourceforge.net/), Trans-Abyss (http://www.bcgsc.ca/platform/bioinfo/software/trans-abyss/releases/0.2) or Oases (http://www.ebi.ac.uk/~zerbino/oases/). Many alternative transcriptome analysis programs now exist. For a survey of different programs, readers may wish to read the study by Garber et al.13 and the review by Martin and Wang14, which describe the comparative advantages and disadvantages and the theoretical considerations of most currently and commonly used programs. The choice of transcriptome analysis strategies (reference-based strategy, de novo strategy, and combined strategy) and programs depends on many factors, including the existence or completeness of a reference genome, the availability of sequencing and computing resources, the type of data set generated and, most importantly, the overarching goal of the sequencing project14.
Another point should be made: computer programs for the transcriptome analysis may not yield a 100% accuracy of all transcript reporting. Sequencing error is another concern. It is always advisable that any differentially expressed transcript identified by RNA-seq be validated by real-time PCR. TaqMan probe-based chemistry is considered the gold standard for real time PCR. Also, newly identified transcripts should be validated by the Sanger method of DNA sequencing before any further experimentation.
Library Preparation: A smear of high molecular weight genomic DNA after the library preparation steps could be caused by a carry-over from the final bead purification step during the library preparation. Returning the libraries to the magnetic stand for a full five minutes may resolve the issue. If not, the issue may stem from incomplete fragmentation of the RNA during the first steps of the library preparation. This could occur if low-quality RNA is used or if the protocol is not followed exactly. For example, altering the incubation times or temperature, adding reagents in the incorrect order or pausing between fragmentation and the synthesis of the first strand of the cDNA. If returning the libraries to the magnetic stand does not remove the high MW DNA, it is advisable to repeat the library preparation with fresh RNA samples.
Low intensity: If the first cycle intensity is lower than expected, it is possible that the primers did not hybridize to the clusters properly during the last step of cluster generation or the flow cell was stored for too long between clustering and the start of the sequencing run. In this case, a primer rehybridization should be performed using a primer rehyb kit from Illumina. Most often, this rehyb solves the intensity issues and the run can be started over without any problem. If, after a rehyb, the intensities are still low, the issue may be with the libraries or the sequence by synthesis (SBS) reagents and Illumina technical service should be contacted for further troubleshooting. If the first read's intensities are good, but the intensities are low after turn around, a primer rehyb of the second read primer may be necessary. This is performed on the sequencing instrument and Illumina technical service should be contacted for the procedure.
High phasing/prephasing: If higher than normal phasing or prephasing is observed, a possible cause is the contamination of the other SBS reagents with cleavage buffer. During the SBS preparation steps, it is essential to handle the cleavage buffer last and change gloves between the handling of the cleavage buffer and any other reagents. If cleavage buffer contamination is suspected, a primer rehyb of the flow cell should be performed and new SBS reagents should be prepared. Alternatively, there could be contamination in the lines of the instrument. If this is suspected, the instrument should undergo a maintenance wash (a water wash followed by a NaOH wash, followed by a final water wash), a primer rehyb should be performed on the flow cell and new SBS reagents should be prepared. If the phasing/prephasing values are moderately high (~0.5), the run may be continued until the Q30 scores and clusters passing filter are calculated. These levels may still be in acceptable limits even with high phasing and/or prephasing.
Generalizability of this protocol
While it is specific to the Illumina HiScanSQ, this protocol is applicable to any of the HiSeq family or the Genome Analyzer II of Illumina instruments (www.illumina.com) with minor modifications of the cluster generation steps and sequencing reagents. Other next-generation DNA sequencing platforms such as the SOLiD series (www.lifetechnologies.com), the GS systems (www.454.com) as well as some emerging newer systems are also employed for the purpose of RNA-seq 11. Although their library construction and sequencing procedures may be slightly different, the RNA handling tips, the data analysis portions (downstream of the CASAVA steps) and the validation by RT-PCR presented in this protocol can be of reference value to their RNA-seq applications.
It should also be pointed out that this protocol is specific to RNA-seq at one time point of thrombin-treated HMVEC-LBl cells, but it could easily be adapted to a multi-time point study or studies in other cells, tissues treated with different stimuli or inhibitors, or comparisons of transcriptomes in cells or tissues between a healthy state and a disease state.
No conflicts of interest declared.
The authors would like to thank Dr. Stephen Kingsmore and the Pediatric Genome Medicine Center at Children's Mercy Hospitals and Clinics for the use of their computing clusters for our data analysis, Illumina's field service team (Elizabeth Boyer, Scott Cook and Mark Cook) and technical consultant team for their quick responses and helpful suggestions on the running of the next generation DNA sequencing instrument, HiScanSQ, and data quality analysis. This work was supported in part by National Institutes of Health Grant HL080042 (to S.Q.Y.) and start-up fund and endowment of Children's Mercy Hospitals and Clinics, University of Missouri at Kansas City (to S.Q.Y.).
|Human Lung Microvascular Endothelial Cells||Lonza||CC-2815|
|Lonza, Bullet Kit||Lonza||CC-3202||Contains EGM-2, FBS, growth factors and antibiotics|
|Ambion mirVana Kit||Life Technologies||AM 1560|
|Experion StdSens RNA||Bio-Rad||700-7103|
|TruSeq RNA Preparation Kit||Illumina||FC-122-1001|
|AMPureXP Beads||Beckman Coulter||A63881|
|Superscript Reverse Transcriptase II||Life Technologies||18064-014|
|Experion DNA 1K||Bio-Rad||700-7107|
|PE Cluster Generation Kit||Illumina||PE-401-3001|
|PhiX Control Kit||Illumina||FC110-301|
|200 Cycle SBS Kit||Illumina||FC-401-3001|
|qPCR machine - Viia7||Life Technologies||Model #VIIA7 / Equipment #10631261||Or equivalent|
|Experion System||Bio Rad||7007001||Bioanalyzer is an alternative system|
|Spectrophotometer||Bio-Tek||Epoch Microplate Spectrophotometer||Or equivalent|
|Centrifuge - Sorvall Legend XTR||Thermo Scientific||75004521||Or equivalent|
|Magnetic stand||Life Technologies||AM10027|
|96-well thermocycler||General Lab Supplier|
|Table 3. List of Key Reagents and Major Equipment. *, In the video, HiSeq1000 instead of HiScanSQ was demonstrated.|
You must be signed in to post a comment. Please sign in or create an account.