Fill out the form to get a free trial:
1Electrical Engineering Department, University of California, Los Angeles, 2Bioengineering Department, University of California, Los Angeles, 3California NanoSystems Institute (CNSI), University of California, Los Angeles
Fill out the form to get a free trial:
We review our recent results on the integration of fluorescent microscopy and imaging flow cytometry tools on a cell-phone using compact and cost-effective opto-fluidic attachments. These cell-phone based micro-analysis devices might be useful for cytometric analysis, such as performing various cell counting tasks as well as for high-throughput screening of e.g., water samples in resource limited settings.
Keywords: Bioengineering, Issue 74, Biomedical Engineering, Medicine, Cellular Biology, Molecular Biology, Electrical Engineering, Telemedicine, Diagnostic Techniques and Procedures, Diagnostic Imaging, Microscopy, Optics and Photonics, Optics, fluorescent microscopy, imaging flow-cytometry, cell-phone microscopy, tele-medicine, global health, wireless health, clinical techniques
Zhu, H., Ozcan, A. Wide-field Fluorescent Microscopy and Fluorescent Imaging Flow Cytometry on a Cell-phone. J. Vis. Exp. (74), e50451, doi:10.3791/50451 (2013).
Fluorescent microscopy and flow cytometry are widely used tools in biomedical research and clinical diagnosis. However these devices are in general relatively bulky and costly, making them less effective in the resource limited settings. To potentially address these limitations, we have recently demonstrated the integration of wide-field fluorescent microscopy and imaging flow cytometry tools on cell-phones using compact, light-weight, and cost-effective opto-fluidic attachments. In our flow cytometry design, fluorescently labeled cells are flushed through a microfluidic channel that is positioned above the existing cell-phone camera unit. Battery powered light-emitting diodes (LEDs) are butt-coupled to the side of this microfluidic chip, which effectively acts as a multi-mode slab waveguide, where the excitation light is guided to uniformly excite the fluorescent targets. The cell-phone camera records a time lapse movie of the fluorescent cells flowing through the microfluidic channel, where the digital frames of this movie are processed to count the number of the labeled cells within the target solution of interest. Using a similar opto-fluidic design, we can also image these fluorescently labeled cells in static mode by e.g. sandwiching the fluorescent particles between two glass slides and capturing their fluorescent images using the cell-phone camera, which can achieve a spatial resolution of e.g. ~ 10 μm over a very large field-of-view of ~ 81 mm2. This cell-phone based fluorescent imaging flow cytometry and microscopy platform might be useful especially in resource limited settings, for e.g. counting of CD4+ T cells toward monitoring of HIV+ patients or for detection of water-borne parasites in drinking water.
Microscopy and flow-cytometry are widely used techniques1-12 in biomedical and scientific research as well as clinical diagnosis for counting and characterization of various cell types. However, conventional microscopes and flow-cytometry instruments are relatively complex and expensive, which limits their use to mainly well-established central laboratories. Recently we have developed a compact and lightweight fluorescent imaging cytometry and microscopy device integrated on a cell-phone,13,14 which shows promise to cost-effectively translate fluorescent microscopy, flow-cytometry and related micro-analysis techniques to resource-limited environments for various telemedicine applications impacting global health.
In the optofluidic flow-cytometry configuration (see e.g. Figure 1C and 1D), a custom-designed polydimethysiloxane (PDMS) based microfluidic channel is positioned in front of the cell-phone camera-unit, where light-emitting-diodes (LEDs) are butt-coupled to the edges of the channel. This microfluidic chip, together with the liquid sample inside, forms an opto-fluidic planar waveguide (composed of for example PDMS-liquid-PDMS) such that the excitation light is guided to uniformly pump the fluorescent labeled specimens inside the micro-channel. The fluorescence emission from these labeled objects, e.g. cells, is further imaged through an additional lens placed right after the cell-phone camera-unit and is mapped onto the cell-phone Complementary Metal-Oxide-Semiconductor (CMOS) image sensor. Since the fluorescent emission is collected perpendicular to the excitation light path, an inexpensive plastic absorption filter is sufficient to remove the scattered excitation light and can provide a decent dark-field background required for fluorescent imaging. Using a similar opto-fluidic design, we can also image the fluorescent objects in static mode (see Figure 1A and 1B), in which the fluorescent particles are sandwiched between two glass slides instead of flowing through a microfluidic channel and the fluorescent emission from these fluorescent particles are captured by the cell-phone CMOS image sensor for particle counting and characterization. Based on different application requirements, flow cytometry or wide-field fluorescent microscopy can be chosen. For example, cell-phone flow cytometry device could be especially useful for screening large volumes of liquid samples (e.g. a few ml) for the detection of rare cells or pathogens.
In this manuscript we review some of our recent results on the integration of fluorescent microscopy and imaging flow cytometry tools on a cell-phone using compact and cost-effective opto-fluidic attachments. These cell-phone-based micro-analysis, imaging cytometry and sensing platforms could provide various opportunities for telemedicine and point-of-care diagnostics, especially impacting our fight against global health challenges in resource limited regions of the world.
In this section, we introduce the experimental protocols for our cell-phone based wide-field fluorescence microscopy13 and opto-fluidic imaging cytometry platform14. We will use fluorescent beads and fluorescently labeled white blood cells to test these imaging platforms.
A. Preparation of Cell-phone Based Wide-field Fluorescent Microscope and Opto-fluidic Imaging Flow Cytometer
The cell-phone based wide-field fluorescent microscope or flow cytometer consists of two major parts: a camera phone and a compact opto-fluidic add-on attachment.
1. The Camera Phone
While the presented techniques are applicable to almost any camera phone, we have chosen Sony Erickson Aino as the base for these devices. This cell-phone has an ~ 8 MegaPixel RGB CMOS sensor installed on it and a built-in lens that has a focal length (f1) of ~ 4.65 mm.
2. Opto-fluidic Attachment for Wide-field Fluorescent Microscopy
The optical attachment is designed by Autodesk and is printed by a Dimension Elite 3-D printer using ABSplus thermoplastic material. In this printing process, model and support materials are heated in an extrusion head within the printer and are deposited layer by layer on a modeling base. When this step is completed, the supporting material can be dissolved, leaving a robust 3D model of the desired prototype Our optical attachment design consists of LEDs (center wavelength at ~ 470 nm, Digikey), a plastic filter (#NT54-46, Edmund Optics), a sample tray, and a plano-convex lens f2=15 mm (# NT45-302, Edmund Optics). All the LEDs and plastic filters can be easily changed based on the fluorophores' spectra. The steps for assembling this opto-fluidic attachment include:
3. Opto-fluidic Attachment for Fluorescent Imaging Cytometry
When there is a need to screen large volumes of liquid samples for the detection of rare events, optofluidic flow cytometry device could be preferred. We can modify our wide-field fluorescent microscope design and convert it into a flow cytometer, where a PDMS based microfluidic channel is used to continually deliver the liquid sample through the imaging volume. The optical attachment is also designed by Autodesk and printed by Dimension Elite 3-D printer. It also consists of LEDs (center wavelength at ~ 470 nm, Digikey), a plastic filter (#NT54-46, Edmund Optics), a sample tray, and an aspherical lens (f = 4.5 mm) (product # C230TME-A; Thorlab). The steps for assembling this opto-fluidic attachment include:
B. Sample Preparation
4. Preparation of Fluorescent Micro-particle Samples
5. Preparation of Fluorescently Labeled White Blood Cells
With our opto-fluidic pumping/excitation scheme (Figures 1C and 1D), fluorescently labeled cells can be continuously delivered into the microfluidic channel using a syringe pump while the cell-phone camera records a time-lapse fluorescent microscopic movie of the flowing cells. These fluorescent movies can then be rapidly analyzed using contour-detection and tracking algorithms14,15 to automatically determine the absolute number and the density of the cells flowing through the microfluidic channels, taking the function of a fluorescent imaging flow-cytometer. Based on the above described platform (Figures 1C and 1D) and sample preparation protocols, we demonstrated accurate counting of total white-blood-cells (WBCs) in human blood samples, achieving comparable results against a standard hematology analyzer.14 Alternatively, we can also image the fluorescent cells/particles in static mode,13 such that without any fluidic flow, achieving a very large sample field-of-view of e.g. ~ 81 mm2 with a spatial resolution of ~10 μm as illustrated in e.g. Figures 2 and 3. Sources of optical aberration can be reduced by designing a more complex lens system. Alternatively, it can also be partially corrected through digital image processing by characterizing the source of the aberration and their spatial patterns.
Figure 1. (A) Schematic illustration and (B) picture of a cell-phone-based wide-field fluorescent microscope.13 (C) Schematic illustration and (D) picture of a cell-phone based imaging flow cytometer. 14 Click here to view larger figure.
Figure 2. The performance of our cell-phone based fluorescent microscope13 is characterized by imaging fluorescent beads (10 μm diameter green and red beads; red bead excitation/emission: 580 nm/605 nm, green bead excitation: 505 nm/515 nm). A decent image quality is achieved over a field-of-view of ~ 81 mm2, see (A,B,C). Toward the edges of this central field-of-view, there are aberrated regions; see e.g. (D-E).
Figure 3. Top image: cell-phone images of fluorescently labeled white blood cells.13 (A-1): Digitally zoomed cell-phone image of fluorescently labeled white blood cells cropped from the top image. (A-2): Compressive decoding results18 for the cell-phone image shown in (A-1). (A-3): Conventional fluorescent microscope (10X objective, NA=0.25) comparison image of the same field of view.
We have presented our recent results on cell-phone based wide-field fluorescent microscopy and opto-fluidic imaging flow cytometry using light-weight and compact opto-fluidic attachments to cell-phone cameras. Using this platform technology we imaged fluorescent objects including micro-particles and labeled white blood cells in whole blood samples. Therefore, this compact and cost-effective cell-phone based fluorescent imaging toolset might be useful for point-of-care diagnosis, especially in resource limited regions for combating various global health problems, such as monitoring of HIV+ patients for their CD4+ T lymphocytes counts.
Besides analysis of bodily fluids, the same cell-phone-based opto-fluidic cytometry platform can also be useful for other sensing needs such as quantification of fluorescent immunoassays for e.g. monitoring of water/food quality in resource-limited settings. Toward this end, we have recently demonstrated sensitive, specific and rapid detection of Escherichia coli O157:H7 (E. coli) in water and milk samples using the same opto-fluidic cell-phone attachment16. We utilized surface functionalized glass capillaries to specifically and sensitively capture E. coli particles in liquid samples that were flown through each capillary. These captured E. coli particles were further labeled with quantum-dot (QD) conjugated secondary antibodies. The fluorescent emission from these functionalized capillaries was then detected using the opto-fluidic cell-phone imaging platform and the integrated fluorescence intensity along the capillary length was used to estimate the density of the captured E. coli particles in the target solution. With this opto-fluidic approach, we demonstrated a detection limit of ~5-10 CFU/ml in both water and milk samples16.
Finally, we should note that depending on the application needs, the optical magnification and resolution of this platform can be tuned by changing f / f2, where f is the focal length of the cell-phone camera lens and f2 is the focal length of the external lens (Figure 1A). In addition to system magnification, we also should note that the LED tray and the associated filter can be easily changed to different colors to accommodate for various fluorophores or even immunochromatographic assays that might be used in specific applications.17
Dr. Ozcan is the founder of a start-up company that aims to commercialize computational imaging and microscopy tools.
A. Ozcan gratefully acknowledges the support of the Presidential Early Career Award for Scientists and Engineers (PECASE), Army Research Office (ARO) Young Investigator Award, National Science Foundation (NSF) CAREER Award, Office of Naval Research (ONR) Young Investigator Award and National Institutes of Health (NIH) Director's New Innovator Award DP2OD006427 from the Office of the Director, National Institutes of Health.
|Cell-phone||Sony||Sony Ericsson Aino|
|Plano-convex lens||Edmund Optics||# NT45-302|
|Aspherical lens||Thorlab||# C230TME-A|
|Polystyrene tube||Fisher Scientific||#05-408-129|
|Red blood cell lysing buffer||Sigma Aldrich||R7757|
|SYLGARD 184 SILICONE ELASTOMER KIT||Dow Corning|
|Red fluorescent beads (10 μm)||Life Technologies||#F8834|
|Green fluorescent beads (10 μm)||Life Technologies||#F8836|
|SYTO16 nucleic acid fluorescent labeling||Life Technologies||# S7578|
You must be signed in to post a comment. Please sign in or create an account.