Enter your email to receive a free trial:
We describe a protocol for the microfabrication of the gradient-generating microfluidic device that can generate spatial and temporal gradients in well-defined microenvironment. In this approach, the gradient-generating microfluidic device can be used to study directed cell migration, embryogenesis, wound healing, and cancer metastasis.
Cite this Article
Chung, B. G., Manbachi, A., Saadi, W., Lin, F., Jeon, N. L., Khademhosseini, A. A Gradient-generating Microfluidic Device for Cell Biology. J. Vis. Exp. (7), e271, doi:10.3791/271 (2007).
A. Microfabrication of the gradient-generating microfluidic device
- The Si wafer is treated with reactive oxygen plasma (5 min at 30W, Harrick Scientific, NY).
- Negative photoresist (SU-8 50, Microchem, MA) is spin-coated at 1000 rpm for 1 min on a Si wafer.
- The wafer is soft baked at 65°C for 10 min and subsequently at 95°C for 30 min on a hotplate.
- The wafer is exposed to UV light (200W) for 3 min through a transparency mask with a minimum feature size of 30 µm.
- The wafer is post baked at 65°C for 1 min and at 95°C for 10 min.
- Si master mold with 100 µm thick channels is developed using SU-8 photoresist developer.
- The wafer containing microchannels is placed in a Petri-dish.
- Poly(dimethylsiloxane) (PDMS) (Sylgard 184) molds are fabricated by mixing silicone elastomer and curing agent (10:1 ratio).
- The PDMS mixture is poured onto the Si master mold.
- The Si master mold is placed on a vacuum desiccator to remove bubbles for 10 min.
- PDMS is cured at 70°C for 1~2 hours.
- PDMS molds are peeled off from the Si master mold.
B. Experimental setup
- Cell inlet, outlet, and infusing inlets of the PDMS-based microfluidic device are punched by using sharp punchers.
- A device and a glass slide (2×3 inch) are irreversibly bonded by the reactive oxygen plasma (5 min at 30W, Harrick Scientific, NY).
- Polyethylene tubing (PE 20, Becton Dickinon, MD) is inserted into the infusing inlets of the microfluidic device and subsequently connected to a syringe pump.
- Fluorescein isothiocyanate (FITC)-dextran (MW=10kD, 10 µM, Sigma) and buffer (PBS, Invitrogen, CA) are infused into the microfluidic device to confirm stable gradients inside the fluidic device.
- Extracellular matrix (ECM) (i.e., fibronectin) is coated inside the microfluidic device for 1 hour in incubator (37°C).
- NIH 3T3 fibroblast cells are trypsinized and dissociated.
- Dissociated cells are loaded into the microfluidic device (800 µm wide) at the cell density of 2×106 cells/ml.
- 2 ml media and 50 ng/ml epidermal growth factor (EGF) is infused into a microfluidic device for generating soluble gradients using a syringe pump (0.05 µl/min).
- Cells are real-time monitored every 5 min by using an inverted microscope (Nikon TE 2000).
Cells exposed to stable concentration gradients of EGF in a microfluidic device migrated toward higher concentrations. The directional orientation of cell migration, chemotactic index, motility of migrating cells were investigated by cell tracking analysis. Therefore, this gradient-generating microfluidic platform could be useful for studying cancer metastasis, embryogenesis, and axon guidance.
|Dextran-FITC||Reagent||Sigma-Aldrich||FD10S||Fluorescein isothiocyanate (FITC) conjugated-dextran (10kD)|
|hr-EGF||Invitrogen||13247-051||human recombinant Epidermal growth factor|
|PDMS||K.R. Anderson Co.||2065622||Poly(dimethylsiloxane) (PDMS), Dow Corning Sylgard 184 (8.6 lb)|
|Negative photoresist||MicroChem Corp.||SU-8 50|
|Si wafer||silicone wafer, 4 inch|
|Polyethylene tubing||BD Biosciences||PE 20|
|NIH 3T3||cell-line||fibroblast cells|
|Inverted microscope||Nikon Instruments||TE 2000|
- Jeon, N. L., Baskaran, H., Dertinger, S. K. W., Whitesides, G. M., Van de Water, L., Toner, M. Neutrophil chemotaxis in linear and complex gradients of interleukin-8 formed in a microfabricated device. Nat. Biotechnol. 20, 826-830 (2002).
- Lin, F., Nguyen, C. M., Wang, S. J., Saadi, W., Gross, S. P., Jeon, N. L. Effective neutrophil chemotaxis is strongly influenced by mean IL-8 concentration. Biochem. Biophys. Res. Commun. 319, 576-581 (2004).
- Chung, B. G., Flanagan, L. A., Rhee, S. W., Schwartz, P. H., Lee, A. P., Monuki, E. S., Jeon, N. L. Human neural stem cell growth and differentiation in a gradient-generating microfluidic device. Lab Chip. 5, 401-406 (2005).
- Saadi, W., Wang, S. J., Lin, F., Jeon, N. L. A parallel-gradient microfluidic chamber for quantitative analysis of breast cancer cell chemotaxis. Biomed. Microdevices. 8, 109-118 (2007).
- Chung, B. G., Park, J. W., Hu, J. S., Huang, C., Monuki, E. S., Jeon, N. L. A hybrid microfluidic-vacuum device for interfacing with conventional cell culture platform. BMC Biotechnol. 7, (2007).
Post a Question / Comment / Request
You must be signed in to post a comment. Please sign in or create an account.